Please click on the following links to download the documents:
ENLock and ENPlus: A New Era of Enteral Nutrition Connections and a Step Forward in Patient Safety
September 2012: As part of a continuous improvement programme and in response to healthcare community requests, Medical Nutrition International Industry (MNI) announces the launch of a new enteral tubing connection system to help avoid accidental misconnections between enteral nutrition and intravenous (IV) delivery systems. ENLock and ENPlus connection systems are enteral nutrition specific and are designed to be incompatible with IV “Luer” based delivery systems, thereby increasing patient safety in the hospital as well as in the home. These connectors resulted from nutrition industry collaboration and will be marketed by enteral nutrition medical device manufacturers in Europe.
The ENLock connection system ensures that only enteral nutrition can be delivered through enteral nutrition delivery systems, and is incompatible with IV system “Luer” connectors: The ENLock connection system includes a connector for enteral administration sets as well as a ‘funnel‘ for feeding tubes such as NasoGastric tubes and PEGs. ENLock syringes and ENLock medication ports prohibit connection from and to IV lines.
The ENPlus connection system ensures that only enteral nutrition feed containers can be attached to enteral feed administration sets and that enteral nutrition administration sets cannot be connected to IV solution containers. Additionally, IV spikes are incompatible with enteral nutrition containers that carry the ENPlus port. The ENPlus connection system includes a plus shaped (+) connector attached to enteral administration sets as well as a corresponding plus shaped (+) port located on feed containers. ENPlus and the current 40 mm Screwcap will co-exist as safe system alternatives.